445 East 77th Street
New York, NY 10075
In the United States there is one FDA approved urinary sphincter available called the Artificial Urinary Sphincter made by Boston Scientific. The sphincter prosthesis is made up of 3 parts: a sphincter cuff, a pump and a reservoir. Everything is located inside the body with no external parts. The sphincter cuff circles around the outside of the urethra and keeps it compressed and closed at rest to prevent urinary leakage.
The cuff is filled with fluid and pressure in the cuff maintained by a fluid filled container which lays in the abdomen called the pressure regulating balloon. A pump placed in the scrotum allows the cuff around the urethra to get deflated and open the urethra up to permit urination.
The sphincter remains closed except when you want to urinate; you squeeze the pump in the scrotum and this transfers fluid out of the cuff and back to the balloon so that you may urinate. The cuff automatically fills up once again after about three minutes closing the urethra and making you continent again.
The operation takes less than two hours to complete and can be done with general or spinal anesthesia – patients often stay usually just overnight. The procedure is usually performed through two small incisions. One, behind your scrotum and another small incision on one side of the lower abdomen. The sphincter is left open and in the “deactivated state” for four to six weeks while patients are healing. During this time In this position incontinence is the same as it was before the surgery. After the four to six week period is up, I activate the device in the office by squeezing the control mechanism near the pump, and usually teach my patients how to use the pump to urinate.
The potential major complications of the prosthesis are urethral atrophy, erosion of the sphincter, infection or a mechanical failure. Urethral atrophy usually occurs over time last the pressure from the cuff on the urethra causes the urethral wall to thin out. As it does this, the pressure decreases on the urethral wall and patients may start to develop incontinence again although usually at a lesser amount. Erosion of the device through the skin occurs very infrequently but occasionally the device can erode through the urethra as well. The symptoms of erosion are usually pain in the penis or scrotum and redness and tenderness over the area and sometimes the prosthesis may be seen through the skin. Erosion through the urethra can manifest as urinary infections or more difficulty urinating. Treatment consists of the administration of antibiotics and removal of that portion of the prosthesis. Infection is a uncommon complication which is more prevalent in diabetics and in people who are undergoing repeat surgery to the same area or in men who have had radiation to the prostate. The most common symptoms of infection is pain over part of the prosthesis, particularly when you are squeezing the control assembly. Less commonly, there may be fever, redness and tenderness over the prosthesis.When infection of the prosthesis is suspected the most conservative treatment is to remove the entire prosthesis surgically. After a suitable waiting period of usually no less than two or three months, a new prosthesis may be implanted.
Mechanical failures of the prosthesis occur in about 10% of people over the course of time. These complications include kinking of the tubing, leaks of the connectors or the cuff itself and malfunctions of the control mechanism. Once a malfunction has been identified, it usually requires a minor operation to replace or fix the malfunctioning part.
Ultimately it is up to you but in my opinion, it remains the gold standards treatment for moderate or severe incontinence. It works well in most patients and if a complication were to arise such as an infection or erosion the device can be easily removed and you still haven’t necessarily burned any bridges for future surgeries in most cases. Most patients are very happy with the artificial urinary sphincter and report dramatic improvements in their quality of life.